Home » Strength of Material Testing Equipment » Pendulum Impact Tester
Strength of Material Testing Equipment
We supply an extensive range of material testing equipment that is fabricated as per the industry defined quality standards and they deliver consistently accurate results. Our range of material testing equipment encompasses equipment for testing & determining deval attrition, dorry abrasion, pendulum impact, tensile strength, thickness, etc. All our tools for material testing are available at industry leading prices and we can execute bulk orders within less lead-time.
Pendulum Impact Tester
The Pendulum Impact Tester is designed for conduction Izod & Charpy Test. The Pendulum is mounted on anti friction bearings. It has two starting positions, the upper one for Charpy and the lower one for Izod testing. On release, the Pendulum swing down to break the specimen and the energy absorbed in doing so is measured as the difference between the height of drop before rupture & height of raise after rapture of the test specimen and is read from the maximum pointer on the dial scale. Charpy Test: The Charpy test piece rests on alloy steel supports anvils fitted on the base of the machine rigidly held in position by alien screws. End stopper is provided for quickly & accurately locating the test piece centrally between the supports. Izod Test The Izod test piece is clamped vertically in Izod support anvils, fitted on the base of the machine. The support is provided with a machined vertical groove to suit the test piece size. The front clamp piece & alien screw enable clamping of the test piece in correct height with the help of Izod setting gauge supplied. Controls of Pendulum: The Pendulum supported in the starting positions is by a self resetting latch. The release mechanism is operated by lever. The Pendulum release latch is operated only when this lever is operated.
Pendulum Impact Testing Machine - Strikers & Supporters
Controls of Pendulum The Pendulum supported in the starting positions is by a self resetting latch. The release mechanism is operated by lever. The Pendulum release latch is operated only when this lever is operated.
Strikers & Supporters There are strikers & one combined support available for fitting into the Pendulum & on to the base of the machine for Izod, Charpy test. Changing from one striker to another is achieved simply by fixing the new striker into its position.